Wewon Environmental Chambers Co., Ltd. is a great dust chamber supplier and made a 3380 liters blowing sand and dust test chamber for a Vietnam client. The controller system of this 3380 litres dust test chamber can be connected to the computer. Meanwhile, The devices has a vacuum pump with a distributing pump or suitable function to keep the talcum powder in suspension.
This 3380 litres dust test chamber made by Wewon Environmental Chambers Co., Ltd. is driven by a fan to blow a particular concentration of sand and dust throughout the test example surface area at a particular circulation price to assess the capacity of these test examples (equipment) subjected to dry sand or a dust-filled environment to shield against the penetration results of dust particles against rough or obstructing impacts of crushed rock and the capacity of the product to shop and run.
The machine has the ability to mimic floating dust, which is primarily made use of to examine the capability of the shell of the checked product to stop the penetration of sand and dust along with the subsequent working performance judgment under the drifting dust condition of the shell and seal of the digital and electrical auto parts.
The Wewon’s 3380 liters blowing dust test chamber style replicates the damaging effect of wind and sand climate on the product, which appropriates for checking the sealing performance of the product’s shell. It is mostly made use of for the IP5X and IP6X tests defined in the enclosure security class criterion.
Just How The dust test Chamber Works? Working Principle of The 3380 Liters Blowing Sand and Dust Test Chamber:
The blowing dust test chamber design has an air circulation with upright blood circulation of dust, and the dust utilized in the test can be reused. The whole air duct is made from imported stainless-steel plate, and the bottom of the air duct is connected with the cone receptacle interface.
The fan inlet and outlet are directly linked to the duct, and then the diffuser on the top of the workshop is attached to the studio body in a suitable placement to form an O-type closed vertical dust blowing cycle system.
Make air flow can stream efficiently, make dust spread uniformly to a big degree. The sand dust test chamber make use of a single large power low noise centrifugal fan, according to the test needs to use variable frequency guv to change the wind speed.